The problem occurred with standard Plainsman M340, M390, M350, M370 and P300. The stonewares have porosities of 2-3%, the M370 1% and the P300 0.5%. Thus, all of these have comfortable margins for overfiring. The G2926B glaze, used on all of them, does seal the surface pretty early so it can contribute to over-fired ware bloating sooner than typical. But the problem here is the cone-fire modes on hobby kiln controllers. For this kiln, the cone 6 program goes to 2236F. That's cone 7. Adding the error of the thermocouple and the misinformation from poorly set cones the temperature overshoot could be more. That means this ware is likely just over-fired. Manually programming your kilns in consort with calibration using self-supporting cones, that is the way to get control. Then you will also be able benefit from the firing schedules like the drop-and-soak PLC6DS and slow-cool C6DHSC.
I was consistently getting the cone on the left when using a custom-programmed firing schedule to 2204F (for cone 6 with ten minute hold). However Orton recommends that the tip of the self supporting cone should be even with the top of the base (they consider the indicating part of the cone to be the part above the base). So I adjusted the program to finish at 2200F and got the cone on the right. But note: This applies to that kiln at that point in time (with that pyrometer and that firing schedule). Our other test kiln bends the cone to 5 o'clock at 2195F. Since kiln controllers fire cone 6 at 2230 (for the built-in one-button firings) your kiln is almost certainly over firing!
I document programs in my account at insight-live.com, then print them out and enter them into the controller. This controller can hold six, it calls them Users. The one I last edited is the one that runs when I press "Start". When I press the "Enter Program" button it asks which User: I key in "2" (for my cone 6 lab tests). It asks how many segments: I press Enter to accept the 3 (remember, I am editing the program). After that it asks questions about each step (rows 2, 3, 4): the Ramp "rA" (degrees F/hr), the Temperature to go to (°F) to and the Hold time in minutes (HLdx). In this program I am heating at 300F/hr to 240F and holding 60 minutes, then 400/hr to 2095 and holding zero minutes, then at 108/hr to 2195 and holding 10 minutes. The last step is to set a temperature where an alarm should start sounding (I set 9999 so it will never sound). When complete it reads "Idle". Then I press the "Start" button to begin. If I want to change it I press the "Stop" button. Those ten other buttons? Don't use them, automatic firing is not accurate. One more thing: If it is not responding to "Enter Program" press the Stop button first.
Put the pots in, select a cone, press start. It is time to rethink that approach! The Bartlett Genesis kiln controller is standard equipment on hobby and production electric kilns now. It is not meant to be run like a toaster! Good glazes are about much more than recipes, they are about firing schedules. None of the built-in "toaster schedules" have hold times on any segments, drop-and-hold sequences or controlled cools. Or even fire-to-cone accuracy. Yet such are a must for defect-free glazes, enhancing the effects of reactive glazes that must develop crystallization or variegation or firing accurately. It is easy to program: Tap the blue edit button to edit a program, tap a column of any segment to edit its value. Tap a segment number to delete or duplicate it. Search "bartlett genesis controller" on YouTube for videos on creating and editing a schedule.
Ware is not turning out as expected and a potter needs to verify the temperature in the kiln. The standard cones on the upper right are misleading. The cone 7 is telling one story but the cone 6 and 5 another. On the lower right is a better way: Self supporting cones. They are always at the right angle and this set of three is bending as expected. To be a full cone 6 the middle one needs to bend just a little more until the tip is even with the top of the base (maybe 2 or 3 degrees). On the top set, the cone 6 is clearly totally flattened and the 5 is a pool of glass, this firing went way beyond cone 6.
Bloating occurs when the off-gassing of decomposing particles in a body has not completed by the onset of density and impermeability associated with the vitrification process.